KROHNE IFC 300 PDF
Electromagnetic flow converter for combination with OPTIFLUX and WATERFLUX flow sensors. KROHNE 03/ – – TD IFC R05 en. The documentation is only complete when used in combination with the relevant documentation for the . 2. The high-performance solution. IFC is the only electromagnetic flow converter with 3 x diagnostics of instrument, application and .
|Country:||Saint Kitts and Nevis|
|Published (Last):||17 August 2017|
|PDF File Size:||9.76 Mb|
|ePub File Size:||15.1 Mb|
|Price:||Free* [*Free Regsitration Required]|
IFC Signal Converter From KROHNE
Group message, when one of the following or some other severe error occurs. Orohne sensor DN [mm] [inch] Min. Available inputs and outputs: In terms of available housing versions, there is a compact variant, in which the signal converter is connected to the measuring sensor, as well as the field, wall-mounted and 19″ rack-mounted housing.
Mounting Of The Compact Version The cables are factory connected and labeled as follows. Application test or operator action necessary. Signaling of the active range possible via a status output adjustable. It is not possible to protect such data completely against access by third parties. However, when other signal cables are used, please note the following electrical data! Only available if “manual” is set in C5.
These include repair, technical support and training. This message is for information only. At the same time, 2 pulse outputs must be available: If this document is not in your native language and if you have any problems understanding the text, we advise you to contact your local office for assistance.
Observe the following notes. Depending on the selection in Kfc. Page 47 Noisy flow indic and flow pipeline is flowing full, C 1.
Availability Of Services The functions of the connection terminals depend on the settings refer to Function tables on page Make sure not kohne damage the stranded drain wires 1, 20, Disposal must be carried out in accordance with legislation applicable in your country.
The change from one measurement value to the next one is limited to the value rkohne limitation” for the total time “pulse krohhe. If you nevertheless use shielded cables, the shield must NOT be connected in the housing of the signal converter.
Products with low conductivity, high solid content or entrained air Inhomogeneous, abrasive and corrosive products Rapid product change Sudden change in pH value Pulsating or turbulent flows IFC Signal Converter.
The manufacturer accepts no liability for the consequences of misuse by the operator. Pipe mounting of the field housing 1 Fix the signal converter to the pipe. The grounding cable should not transmit any interference voltages.
Wrap these around both process connections.
Optionally available virtual reference offers cost savings and increased process reliability as grounding electrodes and rings iff no longer needed. Tighten the screw connection of the cable entry securely.
Depiction via kfc if outputs are held by control input or set to zero. Parameters no longer usable. Check field current connections. Should you nevertheless need to return a device for inspection or repair, please pay strict attention to the following points: Use of the internal power supply: Observe without fail the local occupational health and safety regulations. Other conditions Protection category acc.
Krohne IFC 300 Signal Converter for Electromagnetic Flowmeters
The shielding MUST be connected in the housing of the measuring sensor and signal converter. Aluminium and aluminium sheet, partially polyester-coated Option Versions C and F: Additional grounding instructions are provided in the separate Ex documentation, which are only supplied together with hazardous-duty equipment.
Or the demanded pulse rate is too high.